Electric contact pressure gauges signal too early or too late:
The contact position  is not correct or the contact metal bar is loose. The contact position is not correct and the contact correction is vertical until the appropriate signal occurs. If the metal rod of contact is loose, it should be fixed firmly. If it is slight, it can be used to enlarge the yarn and increase the reverse torque of the wire.
Electrical contact device does not generate signal: contact too dirty contact bad; The insulation layer of the signal device is damped; Circuit failure, etc. If the contact is too dirty, use sandpaper to remove the dirt. When the insulating layer is damp, dry it with hot wind. The circuit is out of order and should be found and repaired.
- What Is A Pressure Gauge
- Pressure Meter Type Selection Princ...
- Measurement Range Pressure Gauge To...
- Considerations For Using Pressure G...
- Pressure Gauge's Main Structure
- Pressure Gauge Installation
- On The Pressure Gauge Upper Limit O...
- Mileage On The Pressure Gauges Sele...
- Pressure Gauge Accuracy
- Quality Requirements For Pressure G...
- The 2018 World Cup Finals
- Voltage Class
- Pressure Gauges Are Mainly Constructed
- Pay Attention To The Use Of Pressur...
- Selection Of Pressure Gauge Type
- The Elastic Pressure Gauge
- Screw Press
- Crank Press
- Multi-station Press
- The Bottom Of The Sea Live